One of the most common questions is understanding the difference of as found versus as left data on a certificate of calibration. First and foremost, these terms apply to data – not to the instrument condition. Many quality professionals get confused because these terms are often used interchangeably. These terms each refer to a specific time when the measurements are recorded. These records are necessary to determine degree of change and for quality tracking.
Defining As Found versus As Left
The as found term refers to measurement data recorded prior to repair or adjustment. This may also be called before data or received data. As found is only used to describe measurements from an item that must be repaired or adjusted to ensure accuracy.
The term as left describes measurement data recorded after repair or adjustment. As left is often used interchangeably with the phrase after data. The as left measurement data provides validation for necessary repairs or adjustments.
A found left term is appropriate when measurement data is recorded for an item that does not require adjustment or repair. The term found left is only used for instruments that passed testing without modification.
The inoperable term is used when an instrument is incapable of being powered on or unusable to test a specific range or function.
Comparing Historical Data
Quality professionals may track calibration data over time for many different reasons. Quality management systems may require a record of test findings or the organization may want to monitor the data trends over time. Managers can use the information to determine if a certain instrument is losing accuracy over a given period or looking at conformance as a whole. This information can help quality departments make decisions about testing or tooling investments.
Example: torque wrench #003 tends to be out of calibration by more than 10% after the most recent test. The Quality Manager checks past history and finds that #003 has been testing out of calibration at a greater rate over the past 3 cycles. This may prompt to manager to shorten the calibration interval or begin planning for replacement of the wrench.
Ultrasonic test sets are used in the aviation industry for ultrasonic inspection. Ultrasonic inspection can be defined as a form of non-destructive testing during maintenance. During ultrasonic inspection, test sets project a high frequency of sound energy to detect flaws and to characterize material of equipment being tested. Ultrasonic test sets can be used to […]
Careless tool control issues cause companies thousands of dollars each year. This is due to tools either being lost, broken or misplaced. Careless tool control issues also include foreign object debris, such as tools being accidentally left within an aircraft after maintenance. In order to avoid these issues from happening, precise tool control is of […]
e2b calibration is an ISO/IEC 17025 ACCREDITED calibration and repair laboratory providing a full scope of NIST TRACEABLE calibration, test, and measurement services for AEROSPACE equipment. Services are provided in a climate-controlled, state-of-the-art laboratory with ON-SITE SERVICES, and local pick-up and delivery available on request. A cloud-based document and ASSET MANAGEMENT PORTAL provides customers with instant access to certificates of calibration, asset information, and reports.